I have recently been using the 2.1 x 150mm 4um 100A Cogent Diamond Hydride™ HPLC Column. I am using a 400 uL/min flow rate with Acetonitrile and an Aqueous Buffer (Column Temperature = 30 °C). I’m seeing a very low back pressure (around 20 bar at the beginning of the gradient while acetonitrile level is high) compared to other HPLC Columns. Is that normal?
After the Water composition Increases over the Gradient, it starts increasing to Back-Pressures that are still low, but more on the usual side of things (40-50 bar, although I am still used to seeing things a bit higher). What kinds of back pressures should I expect for a typical ANP gradient?
Suggestion: The Pressures you describe sound quite typical for these Conditions. You may be used to Reversed Phase Methods that feature higher Water content. Water creates a higher back pressure than Acetonitrile due to lower viscosity of Acetonitrile, so it is normal for the Pressure to be lower than what you may have used in other Methods.
Also temperature increases will Decrease the Pressure as well.