Ferrule Replacement Instructions for the Di-Ad™ HPLC Column Connectors
- Change ferrule after approximately every 50 insertions or when excessive ferrule wear is detected.
- To change ferrule, grasp it in the area of the groove with a pair of wire cutters using minimal pressure and gently pull the ferrule from the threaded stem, wiggling the ferrule if necessary.
- Insert a new ferrule into the threaded stem and screw connector into a column port to seat the ferrule.
The Di-Ad HPLC Column Connectors are designed to provide universal connection to any manufacturer’s 1/16th 10-32 port. Maximum pressure, 6,000 psi.
Click HERE for Di-Ad Ordering Information.